Buy ASTM D R TEST METHOD FOR DETECTION OF HOLES IN MEDICAL GLOVES from SAI Global. This test method is designed to detect holes that leak water and thereby compromise the usefulness of the glove. This test method is suitable and designed as. (Barrier) (ml. GI. ASTM D Water Leak Test). FDA MIL Test Method. ASTM D EN Visual Defects: (i). -Major Visual.
|Published (Last):||20 August 2005|
|PDF File Size:||2.12 Mb|
|ePub File Size:||4.89 Mb|
|Price:||Free* [*Free Regsitration Required]|
A number in parentheses indicates the year of last reapproval.
ASTM D – 06 Standard Test Method for Detection of Holes in Medical Gloves
A superscript epsilon e indicates an editorial change since the last revision or reapproval. This standard has been approved for use by agencies of the Department of Defense.
No other units of measurement are included in this standard. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
The mandrel shall have a clamp or rubber band for securing the glove to the mandrel.
Current edition approved Dec. Originally approved in Last previous edition approved in as D — 99 The markings shall be accurate to within cm3 at volumes greater than cm3.
Affix the glove to the mandrel by stretching the cuff of the glove around the mandrel. A maximum 40 mm of glove cuff should?
Use the securing device, as necessary, to hold the glove in place. The remainder of the glove should hang freely from the mandrel when? The water shall pass freely into the glove.
Let the glove hang for 2 min r5151 again inspect for water leakage. Interpretation of Results 7.
If this leakage occurs, remove and dry the outside of the glove and retest. Precision and Bias 9.
Users of this standard are expressly advised that determination of the validity of any such patent rights, and the risk of infringement of such rights, are entirely their own responsibility. This standard is subject to revision at any time by the responsible technical committee and must be reviewed every?
Your comments are invited either for revision of this standard or for additional standards and should be addressed to ASTM International Headquarters. Your comments will receive careful consideration at a meeting of the responsible technical committee, which you may attend.
If you feel that your comments have not received a fair hearing you should make your d51551 known to the ASTM Committee on Standards, at the address shown below. Individual reprints single or multiple copies of this standard may be obtained by contacting ASTM at the above address or at phonefaxor service astm.