The Pocket Colorimeter II™ has two manuals. The Basic and Full User Manuals can be found below. All Pocket Colorimeter™ II instruments. This symbol, if noted on the instrument, references the instruction manual Hach Pocket Colorimeter™ instruments* are low-cost, high-quality filter photometers. Consult HACH’s Pocket Colorimeter II brochure on ArchiExpo. Page: 1/4.
|Published (Last):||16 December 2017|
|PDF File Size:||20.6 Mb|
|ePub File Size:||19.26 Mb|
|Price:||Free* [*Free Regsitration Required]|
Please contact the Hach Service Center serving your location. Chlorine reagent or one Do not use the same Note Cover with cell holder and cover the uach Shaking dissipates Note: Use a TenSette pipet to add 0.
Order information – Hach-Lange POCKET Colorimeter II User Manual
You can adjust your Community Subscriptions in Settings. Reagents can be relled by contacting Hach or your Hach distributor. Printed in the U. Summary Of Method Summary of Method Chlorine can be present in water as free chlorine and as combined chlorine. Weighing in at only.
Pocekt Certification Hach Company certifies this instrument was tested thoroughly, inspected, and found to meet its published specifications when it was shipped from the factory. Simply zero the instrument with a blank, insert the reacted sample, and read the result.
Order information | Hach-Lange POCKET Colorimeter II User Manual | Page 4 / 4 | Original mode
Chlorine, Free and Total, HR, continued Interferences Interfering Substance Interference Levels and Treatments Monochloramine For conventional free chlorine disinfection beyond the breakpointmonochloramine concentrations are very low.
Determine amount to be added on a separate mL sample, then add the same amount to the sample being tested.
To enter the S0 value, press. Call or E-mail techhelp hach.
Pocket ColoriMeter with standards PCII
The time is entered in hour format. Insert the reagent blank or deionized water into the meter and cover with the cap.
The instrument will MENU use this calibration to determine the displayed concentration of future sample measurements. High range free chlorine determinations are subject to variable levels of interferences from monochloramine. Page of 88 Go. User-Entered Calibration, continued 5. Warranty In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase price, excluding original shipping and handling charges.
Free chlorine is present as hypochlorous acid or hypochlorite ion. Page 88 On the basis of strict liability or under any other legal theory, in no event shall Hach Company be liable for any incidental or consequential damages of any kind for breach of warranty or negligence. For many parameters, the instrument can measure in two different ranges.
Table of Contents Safety Precautions Accurate measurements cannot be obtained unless the sample or blank is covered with the cap. Pokcet and hold again to turn off the backlight.
Pocket Colorimeter II – HACH – PDF Catalogs | Documentation | Brochures
Light emitting diode LED Detector: Error Codes When the instrument cannot perform the function initiated by the operator, an error message will appear in the display. Open the catalog to page 3.
Table of Contents, continued Method Performance Safety Precautions, continued DANGER Indicates a potentially or imminently hazardous situation which, if not avoided, could result in death or serious injury.