Before you When calculating rates.g. Quick Start-Up Guide Setting up an infusion for the Ivac Rate Calculation ml to be infused in 2 hours 30 minutes. The IVAC® Volumetric Pump automatically regulates the infusion rate of intravenous solutions. Users are advised to read, to understand this manual and to be thoroughly familiar with the Pump prior to Tel: (27) (0) interferon α-2α, , , , , , , interferon α-2b, , , , , IVAC, ixabepilone, –,, ,, , ixazomib, .
|Published (Last):||23 April 2014|
|PDF File Size:||16.35 Mb|
|ePub File Size:||17.91 Mb|
|Price:||Free* [*Free Regsitration Required]|
Remove the handset assembly. Front panel labels cannot be reused. Remove the lock nut from the back of the handset inlet assembly and remove the earth braid and shakeproof washer. The live connection is protected by fuse, FS1. Disconnect the flexible circuit from iavc motor by removing the four fixing screws and disconnect de-solder the two motor wires.
Dose limits are not available. Got it, continue to print.
See ‘Basic Features’ for further information. The change of state initiates a countdown to the near end of infusion alarm point in software. Remove 52 syringe shaft assembly from the upper case.
The transmission is not shown here for clarity. F it and fasten the case halves together and check the EOI point again. Note that the pump will require full calibration. Pressure and Max dose icons are shown on the Display.
Lots in Sale Number | Hilditch Group
Where a battery is not tested prior to installation, it is recommended where possible that a battery test is performed. R Remove the rod support retaining screw then remove the circlip and the set cup screw from the latch cam. Alarm Audible alarm Check that the alarm sounds continuously. Syringe constants are also provided for a combination of rates and syringe sizes. To set the hospital name, use the R Q buttons to toggle through characters and the S button to adjust, then select OK to store.
Rzymowskiego 53, Warszawa, Polska. Pump in quiet mode.
Where an item is marked as Fail, the original setting is not changed. The countdown depends upon the syringe type and rate of infusion. I f necessary, separate the lower and upper case of the pump see Chapter 6, ‘Spare Parts Replacement Procedures’ and adjust the position of the EOI actuator on the bottom of the carriage to the desired position, so that the EOI status changes from 0 to 1 at the position set out in table above. Comments to this Manuals Your Name. If this happens, reposition the pump away from the source of interference or turn off the pump and manually regulate the flow.
Item C seal ring is fitted over item D.
Plunger Detect The syringe plunger button position is detected using a slotted optical switch. The display will read EOI opto: Pull the existing 4-way molex header off the board, leaving the pins. B Flexi circuit No manuak A Flexi circuit No 1 Take note of the fold formations and placement of the flexi circuit for reassembly.
CareFusion IVAC 598 Directions For Use Manual
Drugs and protocols use mass dose mode. When servicing, it is recommended that torque is applied gradually until the component is secure. C onnect the Teacher pump to the Learner pump using the RS cable. Linear Speed Sensor The linear movement of the pump transmission is monitored using a slotted optical switch, known as the linear speed opto, coupled with a linear grid.
Disposing of this product correctly will help to save valuable resources and prevent any potential negative effects on human health and the environment which could otherwise arise from inappropriate waste handling. Conditions of Use The Pump should only be operated by a clinician competent in use of automated infusion pumps and post-placement management of intravenous catheters. Refer to the relevant DFU for comprehensive information. Set the oscilloscope range to 0 to 7 VDC. Fit item F – alarm cover label LB to decrease alarm volume if the alarm volume is still too loud at it’s lowest position.
Phyaction Therapy Unit.